Carbon and Sulfur Analyzer (CS20D)
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Transport Package:||Wood Boxes|
|Payment Terms:||T/T, Western Union|
- Model NO.: CS20D
- Carbon 0.05% -6.00%: Sulfur:0.005% -0.300%
- Origin: China
- Type: Element Tester
- Trademark: Joinstars
Supporting the use with the electric arc furnace, capable of rapid and accurate determination of steel, iron, alloy, nonferrous metals, cement, ore, catalysts and other materials of carbon, sulfur elements mass fraction. The device is a set of optical, mechanical and electrical, computer, analytical techniques is one of high-tech products, with a wide measuring range, anti-interference ability, full-featured, easy to operate, accurate and reliable analysis results, etc., and many other industries determination of carbon, sulfur elements ideal analytical equipment. This product is an upgrade of the original CS2D products, increased functionality and reliability, improved cost-effective, the funds invested little user better choice.
The main technical indicators of the technical parameters
1 Analytical methods: Carbon: Sulfur gas volumetric method: Iodometric method
2 Measuring range: 0.05 - 6.00% sulfur carbon: 0.005-0.300%
3 sample weight: 1.0g Measuring Carbon 0.05 to 1.90% sulfur: 0.005-0.060% 0.5g measured carbon: 1.90-3.80% sulfur: 0.060-0.120% 0.25g measured carbon: 3.80-6.00% sulfur: 0.120-0.300%
4 analysis time: With DL1A arc furnace 65 seconds (absorption 85 seconds twice)
5 reading methods: Digital tube direct reading mass fraction, and print the results.
6 analytical error: Carbon to meet the GB/T223.69-97 standard sulfur conform GB/T223.68-97 standard
7 environmental conditions: Room temperature 0-40 ° C, Pressure 90.658 106.658KPa.
Characteristics of the main features:
(1) the gas capacity the differential pressure Dingtan of, the test results by the high sensitivity of the pressure sensor, microcontroller automatic data processing, automated carbon readings.
(2)Iodometric method: Sulfur automatic titration ruled out human error, the analysis of the results of the digital direct reading.
(3)different, said the test results of the sample volume digital direct reading, does not require manual conversion.
(4) sulfur titration plus liquid electrode control, and reduce the failure rate;
(5) The touch buttons isolated to eliminate interference to reduce the failure rate, easy to operate, the new structure.
(6) can choose different materials meeting once or twice to absorb, to improve the accuracy of the analysis.